Document Type dissertation Author Name Chen, Tsai Yuan Email Address tsai125 at gmail.com URN etd-092911-134747 Title Network Electrophysiology Sensor-On-A- Chip Degree PhD Department Electrical & Computer Engineering Advisors John McNeill, Advisor Edward Clancy, Committee Member Michael Coln, Committee Member Keywords EEG ECG EMG Sigma-Delta ADC. Date of Presentation/Defense 2011-04-29 Availability unrestricted
Electroencephalogram (EEG), Electrocardiogram (ECG), and Electromyogram (EMG) bio-potential signals are commonly recorded in clinical practice. Typically, patients are connected to a bulky and mains-powered instrument, which reduces their mobility and creates discomfort. This limits the acquisition time, prevents the continuous monitoring of patients, and can affect the diagnosis of illness. Therefore, there is a great demand for low-power, small-size, and ambulatory bio-potential signal acquisition systems.
Recent work on instrumentation amplifier design for bio-potential signals can be broadly classified as using one or both of two popular techniques: In the first, an AC-coupled signal path with a MOS-Bipolar pseudo resistor is used to obtain a low-frequency cutoff that passes the signal of interest while rejecting large dc offsets. In the second, a chopper stabilization technique is designed to reduce 1/f noise at low frequencies. However, both of these existing techniques lack control of low-frequency cutoff.
This thesis presents the design of a mixed- signal integrated circuit (IC) prototype to provide complete, programmable analog signal conditioning and analog-to-digital conversion of an electrophysiologic signal. A front-end amplifier is designed with low input referred noise of 1 uVrms, and common mode rejection ratio 102 dB. A novel second order sigma-delta analog- to-digital converter (ADC) with a feedback integrator from the sigma-delta output is presented to program the low-frequency cutoff, and to enable wide input common mode range of Â¡Ã“0.3 V. The overall system is implemented in Jazz Semiconductor 0.18 um CMOS technology with power consumption 5.8 mW from Â¡Ã“0.9V power supplies.
Browse by Author | Browse by Department | Search all available ETDs
Questions? Email firstname.lastname@example.org